AGILENT E3640A LABVIEW DRIVER DOWNLOAD
Message 19 of It contains “Getting Started” information on using the driver in a variety of programming environments as well as documentation on IVI and driver specific methods and properties. I did not know this was not accepted and did not plan on playing someone off against someone else,. We fix that by having the software go out and hunt for the instrument on what ever comport it is on. Everything worked the same.
|Date Added:||5 January 2011|
|File Size:||52.90 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Everything f3640a the same. Message 20 of I am meanwhile able to connect to the device via RS and to control it successfully.
Spoiler Highlight to read You are not the first lxbview to forget the front panel setting for external control source! Most Active Software Boards: If you see an error about wscript.
ID Query Manually – 1. During my research online, I have found out that prolific chips have kind of a bad reputation. We have a number of laptops using Lwbview to serial converters to talk to meters and power supplies. These three strings are values of properties in the IIviIdentity interface. This utility does not remove any IVI drivers.
Keysight Technologies / Agilent Technologies hpe364xa Power Supply
Then moved the cable connection to the USB and changed lavbiew Com port ref on the block diag. The current address is displayed momentarily on the front panel when you turn on the power supply. Message 19 of Revision History 1. February 26, Version 1.
Keysight Instrument Drivers | Keysight (formerly Agilent’s Electronic Measurement)
However I found out that I am obviously able to make a connecttion even if I just plug in the converter not linked to the device, so JUST the converter They may not be “broken”, but may not be designed to probvide the correct signal voltage levels for your power supply.
You can download the parser and its associated SDK from: I am however still not able to access the device via USB. Thank you very much for the example for setting agildnt default values myself.
I am also quite sure that I chose the right COM Port, as I always just plug it in and choose the one which newly appeared in the list.
The supported model numbers are: Auto-suggest helps you quickly narrow down your search results by suggesting possible matches as you type. What kind of USB – serial device are you using?
You are not the first person to forget the front panel setting for external control source! You can download the latest version from: Are you saying by “connected” that you are able to send and receive commands when connected to a “built in” comm port of your PC using a program like Terra Term or Hyper Terminal, but not able to send and receive commands when using a 3rd party “RS to USB” converter?
Thanks and sorry again.
Agilent | Agilent E36xx IVI-COM Driver
I fear this might not be fast enough for my system. Is it normal that remote control is that much slower than manual control?
Did you say that you had it communicating once that way? Message 14 of See the section on installation in this document for information on what other software is required by this driver.
That really was the mistake.